This article has multiple issues. Please help improve it or discuss these issues on the talk page. (Learn how and when to remove these template messages)
This article provides insufficient context for those unfamiliar with the subject. Please help improve the article by providing more context for the reader. (October 2009) (Learn how and when to remove this template message)
This article does not cite any sources. Please help improve this article by adding citations to reliable sources. Unsourced material may be challenged and removed.Find sources: "Rigid needle adapter" – news · newspapers · books · scholar · JSTOR (December 2011) (Learn how and when to remove this template message)
(Learn how and when to remove this template message)
Function mode of a rigid needle adapter
A rigid needle adapter enables electrical contact of finest structures on printed and assembled circuit boards, and also direct contact in fine-pole connectors.
The spring probes are arranged in a compact grid in the raster head so that up to 280 spring probes per square centimeter can be integrated. The rigid needles are moved to the desired contact point by the spring probes in the rigid needle adapter. Contacting pitch down to 150 µm can be tested by the movement of the rigid needles.
The contacting stroke is effected in the needle adapter. Contacting forces of 0.4 N to 1.5 N depending on the spring probe are exerted on the contacting point by the spring excursion of approximately 2.5 mm.